|Home||» Products||» Medical Defibrillator||» Refurbished HP Code Master Defibrillator|
HP Code Master Defibrilator (Refurbished)
The HP Code Master that is offered by us are defibrillator, which are obtainable in professionally refurbished with a built-in recorder, ECG, Pacing and AC power with battery back-up. These are available with many functionalities.
Battery Type: Rechargeable, sealed lead-acid battery. 4 Ah, 12 V nominal. Charge Time: 2 hours to 90% of full capacity. 18 hours to 100% capacity. Repeated charging to less than 100% will reduce the useful life of the battery. Capacity: 2.5 hours monitoring or 50 full-energy discharges or 1 hour monitoring and recording. AC input: 100 to 230 VAC +- 15%, 50/60 Hz. Battery Indicators: Illuminated LED indicates battery is charging. Low Battery message appears on the monitor.
Defibrillator Output energy (delivered): 2, 3, 5, 7, 10, 20, 30, 50, 70, 100, 150, 200, 300, and 360 joules. Charge control: push-button on apex paddle and on from panel. Charge time: less than 5 seconds to 360 joules with a functional battery installed. Charge-done indicator: display indicates available energy, charge-done light on apex paddle, charge done tone. Synchronizer: SYNC message appears on the monitor and is annoted periodically on the recorder while in synchronous mode. An audible beep sounds with each detected R-wave, while a marker on the monitor and sync mark on the recorder strip indicate the discharge point. Paddles: standard paddles are anterior / anterior.
Monitor Input: ECG activity may be viewed through the paddles or patient cable. Lead fault: LEADS OFF message and dashed baseline appear on the monitor if a lead becomes disconnected. Common mode rejection: greater than 100dB measured as per AAMI standards for cardiac monitors. Heart rate display: digital read out on monitor from 20 to 280 bpm.
Recorder: Annotatios: time, date, heart rate, event markers, ECG mode, and selected energy. Recorder mode: automatically documents events and ECG during defibrillation episodes. The recorder can be configured can be configured to run in either real time or with a 6-second delay